How do you disinfect an Alger brush?
Completely submerge the device in the cleaning solution and sonicate for at least fifteen (15) minutes. Perform a final thorough rinse of the device and instrument block (if applicable) under running warm tap water for at least one (1) minute. Visually inspect to confirm the removal of debris.
How does Algerbrush work?
It removes the rust ring by brushing of the corneal surface until all rust rings are removed. In this way, the corneal surface is left smooth, which facilitates healing faster and without any residual damage. The alger brush has burrs that are blunt, to avoid further damage or scratching of the corneal surface.